Additional board TYNESS-OPT-PT100
1.Hazards and Warnings
Manufacturer is not responsible for user’s failure to comply with the instructions contained in this manual.
Any service performed on this product must be completed by a qualified individual.
Replacement of this product must be performed by a qualified individual.
Failure to use this equipment in accordance with the specifications in this documentation could lead to a hazard.
No parts in this device should be replaced or removed.
Disconnect all power supplies before servicing the equipment.
This board must exclusively be used with a Tyness product from the company Ewattch.
In the contrasting case, the company Ewattch shall not be held responsible for any deterioration of the equipment.
2.Description and usage
The TYNESS-OPT-PT100 board is an additional board comprising 4 temperature measurement inputs using PT100 3-wire probes.
The board is intended to be set up on our TYNESS sensors (ENERGY and MODULAR).
It comprises 4 terminal blocks of type WAGO series 2059, 3 terminals.
The serigraphy of the printed circuit board specifies the function of the terminal block.
At each transmission, the Tyness returns the instantaneous values of each input.
The board is powered by the Tyness.
Note: reading correction factors cannot be configured in this board.
1 additional standard location for Tyness
4 connectors type WAGO series 2059, 3 terminals
(0.14 to 0.34 mm2- 26 to 22 AWG)
● resolution: 0.01°C
● accuracy: +-0.5°C
● measures: -100°C to +300°C
Number of inputs:
4 PT100 inputs 3 wires
By the TYNESS
Allocation of the terminal blocks:
Pinout of the terminal blocks:
First and foremost, the Tyness and the board must be disconnected.
The board must be inserted carefully and with precision of the connectors on the locations EXT.1 and/or EXT.2 of the Tyness.
The wires of the PT100 probes must be inserted carefully into the terminal
In the case of an extension of the cable of the PT100 probes, we recommend the use of a cable of type SYT1 and AWG24 with rigid core between the “field”
connection terminal block and the “board” terminal blocks.
The board is configured with the Tyness Configurator software.
Always use the latest version of the Tyness Configurator and check that your Tyness has the latest version of the firmware. If necessary, carry out an update.
1 – Click on “Config” of the PT100 board
2 – In the insert “Enabled channels”, define which inputs will be active.
To do so, select:
“Temperature 1” and/or “Temperature 2” and/or “Temperature 3” and/or “Temperature 4”
3 – Define the period of sending of the readings: “Temperature periodicity”
4 – Once all the values are entered, validate by clicking on “Write” If the operation is successful, a window displays “operation successful”.
Validate by clicking on OK
Your PT100 board is ready to operate.
Verification of the readingsIt is possible to display the values of the readings by placing the mouse cursor on the image of the PT100 board.
A Tyness fitted with a PT100 board transmits its data in a raw format on the different LoRaWAN networks. The section below shows you how to decode the frames (Payload) sent by the Tyness.
13, Rue Maurice Jeandon
88100 Saint-Dié des Vosges